The grounding rod seemed to work best when the soil was either damp to very wet. Run each concentration for at least 30 minutes. At 0-ppb, the instrument should within 1 or 2-ppb of the offset value that is set on the D/A board. Conveniently plug it into your nearest 120 VAC outlet — no special wiring needed.
Make sure to determine coverage area from wireless provider and check that service has been activated and SIM card is valid. Solution: Make sure power is on to modem and with the correct polarity. Set the switch next to the thumbwheel to ‘Auto’ Note: Make sure to allow each instrument being calibrated to run for at least an hour so the lamp and other elements inside reach operational temperatures. Problem: Errors were received while trying to send script file. Construct a 9 pin serial cable available that is connected one-to-one for each pin with a female plug on one end and a male plug on the other.
Grid current, plate current and forward PEP output power are continuously monitored to tell you of improper loading and abnormal conditions. Select the file and follow the prompts. As each command is executed, an “OK” response should appear. The message “Enfora, L.P.” should appear followed by an “OK.” Type the following commands and press enter. An “OK” should appear after each command is executed. Enter the network address such as 10.200.4.XX where XX is the number assigned to the modem in the script. If the Zeno 3200 responds then the modem should be ready to go.